Reliable, highly accurate measurements for the test and development of power electronics.
The compact Fluke Norma Series Power Analyzers provide the latest measurement technology to assist engineers with the development and testing of motors, inverters, lighting, power supplies, transformers and automotive components.
Based on a patented, highbandwidth architecture, the instruments deliver highprecision measurements of single or threephase current and voltage, harmonics analysis, Fast Fourier Transformation (FFT) analysis, as well as calculations of power and other derived values.
The Fluke Norma 5000 SixPhase Power Analyzer is a rugged, highprecision analyzer providing unmatched price performance for easy and reliable use in the field, or as a bench unit in laboratories and on test benches.
Applications
Other useful features:
General Specifications  
Number of Phases  1 to 3  
Weight  Approx. 5 kg (11 lb)  
Size  150 x 237 x 315 mm  
5.9 x 9.3 x 12.4 in  
Onboard Printer  No  
Display  Color, 5.7 in / 144 mm  320 x 240 pixel  
Userselectable background lighting and contrast.  
Bandwidth  DC to 3 MHz or DC to 10 MHz depending on input module  
Basic Accuracy  0.2%, 0.1% or 0.03% depending on input modules  
Sampling Rate  0.33 MHz or 1 MHz depending on input modules  
Voltage Input Range  0.3 V to 1000 V  
Current Input Range (direct, not via shunt)  0.03 mA to 20 A depending on input module  
Memory for Configurations  4 MB  
Memory for Settings  0.5 MB  
Fast Fourier Transformation (FFT)  To the 40th harmonic  
RS232/USB Interface  Standard  
PI1 Process interface (8 analog/impulse inputs and 4 analog outputs)  Optional  
IEEE 488.2/GPIB interface (1 MBit/s Ethernet / 10 MBit/s or 100 MBit/s)  Optional  
Fluke NormaView PC software (for data download, analysis and report writing)  Standard  
Basic Functions  
Fast Fourier Transformation (FFT)  Calculation of harmonics with graphical representation. Up to 3 bar graphs are displayed at the same time.  
Measured values: U, I, P per phase  
Order: 1st to 40th harmonics, maximum half sample frequency  
Digital Oscilloscope (DSO)  Simultaneous display of up to 3 measured values on sample level. Quick view of curve form and distortion.  
Integration function (energy)  Simultaneous display of up to 6 configurable numeric values. Start/Stop conditions and positive negative direction available.  
Vector Display  Vector display of HO1 up to 6 signals. For easy testing of the right connection of the instrument and quick overview of the phase angle of each signal.  
Recorder  Display of average values over time for trend determination.  
RAM data memory  Storing of sample and average values; setting of start and stop conditions.  
From the RAM approximately 4 MB are available for the storage of measured values.  
Configuration  Set up the analyzer to measure and display data in the format required.  
Ambient Conditions  
Working Temperature Range  5°C to 35°C (41°F to 95°F)  
Storage Temperature Range  20°C to 50°C (4°F to 122°F)  
Housing Material  Fluke Norma Power Analyzers are extremely compact and equipped with a solid metal case to meet stringent EMC requirements.  
Climatic Class  KYG DIN 40040, max. 85% relative humidity, noncondensing.  
Power supply  85 V AC to 264 V AC, 50 Hz to 60 Hz, DC 100 to 260 V, ca. 40 VA European plug with switch. Binding post for current available on some models.  
Measuring Inputs  Safety sockets 4 mm, 2 for each input. External shunt connection over BNC socket.  
Operation  Membrane keyboard with cursor – function keys and direct functions.  
Connections  Rear panel of the 3phase Analyzer  
Measured Values  
Nongapping calculation of averaged values for each phase. In three phase system additionally calculation of total power and averaging of V and I of the three phases. The fundamental H01 will be calculated in synchronous mode also for these values.


Frequency and Synchronization  
Range  DC and 0.2 Hz to sample rate  
Accuracy  ±0.01% of measured value (reading)  


Configuration Memory  
Up to 15 user configurations can be saved into a permanent memory and reloaded later on. Changes that were not saved are lost after switching off the instrument.  
Interface  
RS232 interface for upload of firmware and data exchange with the PC. A printer can be connected over an external converter.  
Options  IEEE 488.2 / 1 MBit/s  
Ethernet / 10 MBit/s or 100 Mbit/s  
Standards and Safety  
Electrical Safety  EN 610101 / 2nd Edition 1000 V CAT II (600V CAT III)  
Degree of pollution 2, safety Class I  
EN 61558 for transformer  
EN 610102031/032 for accessories  
Maximum inputs  For voltage inputs Measurement range 1000 Veff, 2 kVpeak  
For current inputs Measurement range 10 Aeff, 20 Apeak  
Test voltage  Net input  case (protective conductor): 1.5 KV AC 
Net connection  Measurement input: 5.4 kV AC  
Measurement inputs  Case: 3.3 kV AC  
Measurement input  input: 5.4 kV  
Electromagnetic susceptibility  Emission  IEC 613261, EN 500811, EN 55011 Class B 
Immunity  IEC 613261 / Annex A (industrial sector), EN 500821 
Power Phases
The Fluke Norma 4000 Power Analyzer can be equipped with up to three power phases. Users can select the power phase best suited for their application.
Each modular plugin power phase consists of a voltage and a current measurement channel. Each measuring channel is available for each basic unit, however only one kind of channel can be used per unit. Check standard configurations.
Power Phase Overview
Power Phase Channels  
PP42  Accuracy  0.2% (0.1% rd + 0.1% rg) 
Current range  20 A  
Sampling rate  341 kHz  
Bandwidth  3 MHz  
PP50  Accuracy  0.1% (0.05% rd + 0.05% rg) 
Current range  10 A  
Sampling rate  1 MHz  
Bandwidth  10 MHz  
PP54  Accuracy  0.1% (0.05% rd + 0.05% rg) 
Current range  10 A  
Sampling rate  341 kHz  
Bandwidth  3 MHz  
PP64  Accuracy  0.03% (0.02% rg + 0.01% rg) 
Current range  10 A  
Sampling rate  341 kHz  
Bandwidth  3 MHz 
Current Shunts
Current Shunts
Planarshunt  
Continuous load (Imin) Range (Imax) 
0.3 A 
32 A  
Nominal current (calibration point)  32 A 
Basic accuracy at calibration point [%]  ± 0.03 
Basic accuracy at nominal current [%]  ± 0.03 
Nominal voltage drop [V]  320 mV 
Nominal resistance [Ohm]  10 mΩ 
Short time overrange (5s load 15s interval) 
100A/ 1s5s 
Overload Wmax   
Bandwidth  DC  1 MHz 
Frequency range   
Angular accuracy [°/kHz]  ± 0.1 
Frequency influence [%/kHz]   
Load influence [%/A2]   
Temperature coefficient [ppm/K]  ≤10 
Weight (kg / lbs)  0.62 /1.36 
Triaxialshunts  
Continuous load (Imin) Range (Imax) 
0.1 A  1 A  6 A 
30 A  100 A  300 A  
Nominal current (calibration point  10 A  30 A  100 A 
Basic accuracy at calibration point [%]  ± 0.03  ± 0.03  ± 0.1 
Basic accuracy at nominal current [%]  ± 0.03  ± 0.03  ± 0.1 
Nominal voltage drop [V]  100 mV  30 mV  20 mV 
Nominal resistance [Ohm]  10 mΩ  1 mΩ  0.2 mΩ 
Short time overrange (5s load 15s interval)  35 A  200 A  450 A 
Overload Wmax  90 W  200 W  2 kW 
Bandwidth  2 MHz  2 MHz  1 MHz 
Frequency range  0 to 500 kHz  0 to 200 kHz  0 to 100 kHz 
Angular accuracy [°/kHz]  ± 0.001  ± 0.002  ± 0.002 
Frequency influence [%/kHz]  ± 0.0015  ± 0.0015  ± 0.01 
Load influence [%/A2]  1 x 10^{6}  1 x 10^{6}  1 x 10^{6} 
Temperature coefficient [ppm/K]  ≤15  ≤15  ≤10 
Weight (kg / lbs)  0.75 / 1.65  0.75 / 1.65  1.2 / 2.65 
Triaxialshunts (Cont.)  
Continuous load (Imin) Range (Imax) 
2 A  18 A  18 A 
450 A  1000 A  1500 A  
Nominal current (calibration point  150 A  300 A  500 A 
Basic accuracy at calibration point [%]  ± 0.1  ± 0.1  ± 0.1 
Basic accuracy at nominal current [%]  ± 0.1  ± 0.1  ± 0.1 
Nominal voltage drop [V]  75 mV  15 mV  30 mV 
Nominal resistance [Ohm]  0.5 mΩ  0.06 mΩ  0.06 mΩ 
Short time overrange (5s load 15s interval)  650 A  1500 A  2000 A 
Overload Wmax  10 kW  7.5 kW  10 kW 
Bandwidth  DC  500 MHz  DC  500 kHz  DC  200 kHz 
Frequency range  0 to 100 kHz  0 to 20 kHz  0 to 20 kHz 
Angular accuracy [°/kHz]  ± 0.025  ± 0.025  ± 0.025 
Frequency influence [%/kHz]  ± 0.03  ± 0.03  ± 0.03 
Load influence [%/A2]  0.5 x 10^{6}  0.2 x 10^{6}  0.5 x 10^{6} 
Temperature coefficient [ppm/K]  ≤10  ≤10  ≤10 
Weight (kg / lbs)  6 / 13.2  5.3 / 11.7  6 / 13.2 