Update Terakhir

POWER METER PW3336

Negara Asal

JEPANG

Pembelian minimum

1

Harga

Kategori

2-Channel AC/DC Power Meter for Single-Phase to 3P3W Equipment such as Motors, Inverters, Power Conditioners, and Power Supplies

Hioki benchtop power meters and power analyzers are best in class power measuring instruments for measuring single to three-phase lines with a high degree of precision and accuracy. The PW3336 series measure power on the full range of electrical equipment, from single-phase devices such as battery-driven devices and household electronics to industrial use and three-phase electrical equipment over 2 input channels. The power meters provide coverage from DC to inverter frequencies with support for voltages up to 1000 V and direct-input currents of up to 65 A, and can also measure currents of up to 5000 A using an optional current sensor.

**Compatible with the SPECpower® (*1) benchmark for server power consumption****Measure DC and single-phase 2-wire to 3-phase 3-wire with 2-channel input**

**For development and production of motors, inverters, power conditioners, power supplies, and other devices**

**High-precision basic accuracy of ±0.1 % (*2)**

**Wide frequency bandwidth of 0.1 Hz to 100 kHz or DC**

**High-current measurement up to 65 A of direct input**

**Harmonic measurement up to the 50th order according to IEC 61000-4-7**

**High-accuracy measurement, even with a low power factor for no-load testing of transformers and motors**

**Built-in external sensor input terminals to measure up to 5000 A AC**

**Synchronize up to 8 units for multi-unit measurement**

- *1:SPECpower® is a registered trademark of Standard Performance Evaluation Corporation.
- *2:For complete details, please refer to the specifications.

Accuracy guaranteed: 1 year

Measurement lines | Single-phase 2-wires, single-phase 3-wires, 3-phase 3-wires, (voltage / current measurement range set for each wiring mode) | |||||
---|---|---|---|---|---|---|

Measurement items | Voltage, Current, Active power, Apparent power, Reactive power, Power factor, Phase angle, Frequency, Efficiency, Current integration, Active power integration, Integrated time, Voltage waveform peak value, Current waveform peak value, Voltage crest factor, Current crest factor, Time average current, Time average active power, Voltage ripple factor, Current ripple factor | |||||

Harmonic parameters | Synchronization frequency range: 10 Hz to 640 Hz, Analysis order up to 50th Harmonic voltage RMS value, Harmonic current RMS value, Harmonic active power, Total harmonic voltage distortion, Total harmonic current distortion, Voltage fundamental waveform, Current fundamental waveform, Active power fundamental waveform, Apparent power fundamental waveform, Reactive power fundamental waveform, Power factor fundamental waveform (displacement power factor), Voltage current phase difference fundamental waveform, Interchannel voltage fundamental wave phase difference, Interchannel current fundamental wave phase difference, Harmonic voltage content % , Harmonic current content % , Harmonic active power content % (The following parameters can be downloaded as data during PC communication but not displayed: Harmonic voltage phase angle, Harmonic current phase angle, Harmonic voltage current phase difference) |
|||||

Measurement range (*1) | [Voltage] 0.15 V to 1000 V AC/DC
[Current] |
|||||

Integration measurement (Integration time up to 10,000 hours) |
[Current] No.of displayed digits: 6 digits (from 0.00000 mAh, Polarity-independent integration and Sum value) [Active power] No.of displayed digits: 6 digits (from 0.00000 mWh, Polarity-independent integration and Sum value) |
|||||

Input resistance (50/60 Hz) | [Voltage] 2 MΩ, [Current] 1 mΩ or less (direct input) | |||||

Basic accuracy (Active power) |
±0.1% rdg. ±0.1% f.s. (DC) ±0.1% rdg. ±0.05% f.s. (45 Hz to 66 Hz, at Input < 50% f.s.) ±0.15% rdg. (45 Hz to 66 Hz, at 50% f.s. ≤ Input) |
|||||

Display refresh rate | 5 times/s to 20 seconds (depend on average times settings) | |||||

Frequency characteristics | DC, 0.1 Hz to 100 kHz | |||||

D/A output (-02/-03 model only) |
16 channels (selectable from following items), Level output DC ±2 V, Waveform output 1 V f.s. Level output, instantaneous waveform output (voltage, current, active power) Level output (apparent power, reactive power, power factor, or other) High-speed active power level output |
|||||

Functions | [Rectification method] AC+DC, AC+DC Umn, AC, DC, FND, Auto-range, Average, VT or CT ratio settings, Synchronized control, MAX/MIN, or other functions | |||||

Interfaces | RS-232C / LAN standard, (-01/-03 model also includes GP-IB) | |||||

Power supply | 100 to 240 V AC, 50/60 Hz, 40 VA max. | |||||

Dimensions and mass | 305 mm (12.01 in) W × 132 mm (5.20 in) H × 256 mm (10.08 in) D, 5.2 kg (183.4 oz) | |||||

Included accessories | Instruction manual ×1, Measurement guide ×1, Power cord ×1 |

- *1:MIN./MAX. current values and accuracy will vary depending on the current sensor used.

3NH
AccuTrak
AELAB
AEMC
AIRMAR
AMPROBE
APERA INSTRUMENTS
AS ONE
ASTM HYDROMETER
ATAGO
B-ONE
BACHARACH
Bahan Kimia
BENCHMARK
BINDER
BOECO
Boekel Sci
Cannon Instrument
Cytiva
DLAB
EDC
EUTECH
EXTECH
Felix Instrument
FLUKE
FRITSCH
GLASSWARE
HACH
HANNA Inst.
Heidolph
HIOKI
HYDRAJAWS
HYDRANAL
IKA
JENWAY
JOANLAB
JULABO
Koehler Instrument
KRUSS
KYORITSU
LAUDA
LOVIBOND
Ludwig Schneider
LUTRON
Marvel Coolbox
MEMMERT
METTLER TOLEDO
MITECH
OHAUS
PCE
PERKIN ELMER
PIPETUS
PSL RHEOTEK
RETSCH
ROBINAIR
ROBINSON MANUFACTURING
SAUERMANN
SCHALLER HUMIMETER
SHINAGAWA
Simple SDI
SK SATO
STUART
TELEDYNE FLIR
TEXIM
THERMO SCIENTIFIC
TROTEC